4U4022 LITE Liner Silikon SkinTex 2 SOFT Locking
Mobility class: 1-3
for BK stumps
Connector: Locking (with distal connector)
- made from medical grade* silicone gel
- seamless fabric cover
- standard with approx. 10 cm matrix
- high comfort by using approx. 2,5 mm ComfortSil Silicone Gel
- high adhesion between liner and skin
- perfect pressure distribution
- textile fabric for easy donning
- reduced pistoning by the matrix
- suitable for nearly all locking devices
|Article number||Matrix length||Distal Circumference|
|4U4022-10-18||approx. 10 cm||18 cm|
|4U4022-10-20||approx. 10 cm||20 cm|
|4U4022-10-21||approx. 10 cm||21 cm|
|4U4022-10-22||approx. 10 cm||22 cm|
|4U4022-10-23,5||approx. 10 cm||23,5 cm|
|4U4022-10-25||approx. 10 cm||25 cm|
|4U4022-10-26,5||approx. 10 cm||26,5 cm|
|4U4022-10-28||approx. 10 cm||28 cm|
|4U4022-10-30||approx. 10 cm||30 cm|
|4U4022-10-32||approx. 10 cm||32 cm|
|4U4022-10-34||approx. 10 cm||34 cm|
approx. 10 cm
|4U4022-10-38||approx. 10 cm||38 cm|
|4U4022-10-40||approx. 10 cm||40 cm|
|4U4022-10-42||approx. 10 cm||42 cm|
|4U4022-10-45||approx. 10 cm||45 cm|
How to choose the correct liner size
a) Measure the circumference of the residual limb at 4 cm from the distal end.
b) Choose the liner size corresponding to this measurement or the next smaller size, respectively.
c) Do not choose a liner whose size exceeds the measurement taken.
d) Never reduce by more than one liner size in order to avoid unnecessarily high compression.
In locking liners, a matrix with 10 cm length is integrated as a standard.This matrix should end below the head of the fibula and must not exceed it, as this could result in problems during knee flexion.