LITE Liner® Silicone
The connection between residual limb and socket is the key for a real high-quality fitting of a prosthesis.
The wide range of silicone liners eased it to find the best solution for your patient upon the indication for the different needs.
LITE Liner® Silicone features a high stability and a moderate damping. They are available with or without distal connector and in different sizes and colours.
LITE Liner® Clear
LITE Liner® Skin Tex Seam (two-parted fabric)
LITE Liner® Skin Tex Seamless (no seam)
|approx. 2,5 mm DuraSil-Silikon, hardness: normal
approx. 2,5 mm ComfortSil-Silikon, hardness: soft
approx. 2,5 mm Duosil-Silikon, Duosil
POST-OP Liner, very soft, no fabric cover
approx. 6 mm ComfortSil-Silikon, hardness: soft
approx. 5 cm Matrix
approx. 10 cm Matrix
approx. 15 cm Matrix
total length Matrix (for AK), thickness approx. 2,5 mm
|LITE Liner® Clear
LITE Liner® SkinTex
LITE Liner® SkinTex, no seam
|blue (only LITE Liner® SkinTex Seam)
light grey (LITE Liner® SkinTex Seam & Seamless)
clear (only LITE Liner® Clear)
skin coloured (only LITE Liner® Clear)
|Ordering example: 4W2022-10-28-G = LITE Liner® Skin Tex 2 soft, 10 cm Matrix, size 28, Color light grey|
Determination of the matrix length:
For different lengths of the residual limb, the length of the matrix (reinforcement) can be chosen from 0, 5, 10 and 15 cm. The matrix should cover the tibia but should not be extended above the level of the fibula head. This ensures a higher stability of the soft tissue.
Determination of the size:
To determine the correct liner size, the circumference of the residual limb is measured 4 cm above the distal end. Select that size being exactly the same of the stump circumference or the next smaller one.
If the profile of the residual limb is to be compensated, the Distal Cup is recommended for conical stumps making it more cylindrical. The size of the silicone liner is then to be determined above the donned Distal Cup.
In case of a flat or square formed residual limb, the Silicone Pad can be used to replace missing tissue and to achieve an optimum contact between skin and silicone.
The silicones used have been tested for biocompatibility according to DIN EN ISO 10993-5. The certificates can be delivered anytime on request.